BS EN ISO 6414:2020 pdf – Technical product documentation (TPD) — Technical drawings for glassware.
1 Scope This document establishes rules and conventions for particular use with technical drawings on glassware, for example, laboratory glassware or glassware used in other technical fields. Optical parts are not, however, included herein. 2 Normative references The following documents are referred to in the text in such a way that some or all of their content constitutes requirements of this document. For dated references, only the edition cited applies. For undated references, the latest edition of the referenced document (including any amendments) applies. ISO 128-34:2001, Technical product drawings — General principles of presentation — Part 34: Views on mechanical engineering drawings ISO 128-50:2001, Technical drawings — General principles of presentation — Part 50: Basic conventions for representing areas on cuts and sections ISO 129-1:2018, Technical product documentation (TPD) — Presentation of dimensions and tolerances — Part 1: General principles ISO 383, Laboratory glassware — Interchangeable conical ground joints ISO 641, Laboratory glassware — Interchangeable spherical ground joints ISO 4793, Laboratory sintered (fritted) filters — Porosity grading, classification and designation ISO 10209, Technical product documentation — Vocabulary — Terms relating to technical drawings, product definition and related documentation 3? Terms? and? definitions For the purposes of this document, the terms and definitions given in ISO 10209 apply. ISO and IEC maintain terminological databases for use in standardization at the following addresses: — ISO Online browsing platform: available at https://www.iso.org/obp — IEC Electropedia: available at http://www.electropedia.org/
4 General 4.1 As a general principle, all glassware shall be drawn as if it were non-transparent (opaque), according to ISO 128-34:2001, Clause 15. 4.2 In terms of observation of calibration, font and pointer, the outer glassware shall be drawn as a transparency.
7 Thin walls 7.1 Drawing of wall thickness When drawn in section, thin walls shall be represented, in spite of their real wall thickness, by lines with a thickness of at least twice that used for visible outlines (see Figures 5, 6, 7, 8, 9 and 10).
6 Special glassware indications 6.1 Interchangeable conical ground joint Interchangeable conical ground joints conforming with the requirements of ISO 383 shall be designated in the manner described therein. Accordingly, no detailed dimensioning of that portion and no indication of the surface finish are required. An example of a code identification for interchangeable conical ground joints is shown in Figure 3.